Features and Functions Function Limitation If you change to a function where the maximum frequency is less than the current function, the frequency will be adjusted to the maximum value for the new function. If there are any missing or mechanically damaged items, contact the nearest Agilent Sales Office. Features and Functions 8 Saving the waveform. Click Arb to select the Arbitrary function. Always refer to Agilent web site for latest revision. In FM, the frequency of the carrier is varied by the instantaneous voltage of the modulating waveform. For example, if you are currently outputting a 20 MHz sine wave and then change to the Ramp function, the UA will automatically adjusts the output frequency to kHz the upper limit for Ramp.
|Date Added:||8 September 2014|
|File Size:||52.78 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
At power-on, the units for output amplitude are volts peak-to-peak. A Pass – Normal operation, no effect.
AGILENT TECHNOLOGIES AGILENT UA USER MANUAL Pdf Download.
Check Your System Prior to any installation or configuration, please ensure that your PC meets the following minimum system requirements. Agilent 33250a labview represents the amount of time per cycle that the Ramp wave is rising assuming agilent 33250a labview the waveform is not inverted.
Page 32 Page 33 – E. In AM, the amplitude of the carrier is varied by the instantaneous voltage of the modulating waveform. For a linear sweep, the UA varies the labviww frequency in a agilent 33250a labview fashion during the sweep. To output afilent PSK waveform, configure the settings for the carrier frequency, output amplitude, offset voltage, deviation, and PSK rate. Got it, continue to print.
Arbitrary waveform using the soft front panel. Features and Functions 8 Saving the waveform.
Click Arb to select the Arbitrary function. Analog signal generator vector signal generator 66 pages. Click to begin the freehand draw mode or enable line draw mode. The maximum carrier frequency depends on the selected function as shown below. The default is 1 kHz for all agilent 33250a labview.
Output Agilent 33250a labview You can disable or enable the soft front agilent 33250a labview output control. Offset Agildnt Limitations The relation between the output amplitude and offset voltage is shown below. Page Index limits amplitude duty cycle frequency function offset voltage output termination units selection L-Mount kit output amplitude control frequency function SCPI commands sweep termination units phase noise carrier frequency carrier waveform modulating waveform modulating waveform frequency phase deviation SCPI commands prerequisites product dimensions The default PSK rate value is 10 Hz.
The following SCPI commands show a sample procedure of generating output. For example, if you are currently outputting a 20 MHz sine wave and then change to the Ramp function, the UA agilent 33250a labview automatically adjusts the output frequency to kHz the upper limit for Ramp. Check Your System Page 29 – B. Don’t show me this message again.
The Driver Signing Options dialog box will appear. This is the default trigger source for both soft front panel and remote interface use. If the actual agilent 33250a labview impedance is different from the specified value, the agilent 33250a labview and offset levels will be incorrect.
Features and Functions Soft Front Panel Operation On the Offset panel as shown in desired offset value and select the unit from the drop down list.
Getting Started If you do not wish to receive similar warning messages in the future, N O T E follow the instructions below. Follow the instructions on the screen to proceed with the installation. Select Period agilent 33250a labview use the numeric keypad or numeric drop down waveform. To output the 3320a waveform, configure the agilent 33250a labview for the carrier frequency, output amplitude, offset voltage, modulating frequency, phase deviation, and the desired waveform.
Agilent Technologies AGILENT U2761A User Manual
Features and Functions Modulating Waveform Soft Front Panel Operation On the Frequency panel as shown in desired frequency value and select the unit from the drop down list.
Install the Module Driver Page 31 – D. Agilent 33250a labview disable this dialog, select the Never show this dialog option in the Show this dialog panel and click 9 For modules other than the UA Series, UA Series, UA Series, agilent 33250a labview UA, the system will perform a firmware version check on your connected